AngelMed Guardian® Case Study
Caution: Investigational device. Limited by
United States law to investigational use.
History – Multivascular CAD, hypertention, hyperlipoproteinemia, unstable angina, and diabetes (Type II). Patient has a family history of premature heart disease, sustained a prior heart attack, and received a stent in the ostium of the RCA. Current medications include clopidogrel, lovastatin, aspirin, fenofibrate, beta blocker (nebivolol), Trilipix, triamterene/HCTZ, nitroglycerin prn. Additional medications for the treatment of diabetes.
Doctors originally implanted the Guardian on 29 Nov 2007. There were no alerts for 3.5 years until the implant issued a See Doctor alert to replace the device due to battery depletion. On 5 Jan 2011, doctors replaced the device and discharged the patient without incident.
While teaching on the morning of 16 Aug 2012, the patient’s
Guardian device issued an Emergency alarm. Rather than call for
an ambulance, the patient opted for a family member to take her to
the hospital. At the time of the alarm, she was asymptomatic, but
experienced some back pain and nausea on her trip to the hospital.
Upon arrival, the attending physician ordered standard ECG and
blood tests. The ECG was negative as were the blood enzyme
levels (Troponin 0.01, CK-MB 1.9, Total CPK 88) - demonstrating
no indication of cardiac injury or necrosis.
Doctors retrieved data from the Guardian IMD, which revealed that the Guardian device detected an acute ischemic event at the time of the alarm. As a consequence, doctors admitted the patient and scheduled her for a cardiac catheterization the following day.
Explanation of Guardian Data:
The Guardian data, shown in Figure A, demonstrate that at 8:22am
a positive ST Shift event occurred while the patient’s heart rate was
in the normal range, indicating an acute ischemic event. The ST
Shift was measured to be 27.4%, which exceeded the 25%
ischemia threshold and was significantly greater than the typical ST
Shift variation for this patient, which ranged between -4% and +6%.
Figure A -
ST Shift and Heart Rate at Alarm
The initial ST Shift spike lasted about 1.5 minutes before moderating to still-elevated levels. Twenty-two minutes later, the ST Shift level spiked again briefly above the ischemia threshold and remained at elevated levels for over an hour.
Figure B -
Figure C -
Emergency Alarm ECG
Figure B shows an electrogram of the hourly baseline in effect at the time of the alarm. Figure C shows the electrogram that triggered the alarm. The advancement of the T-wave raises the average voltage value of the ST segment enough to exceed the patient’s ischemia threshold and cause the alarm.
The catheterization identified: a 90% focal stenosis of the previously placed RCA stent several de novo lesions in the LCX, resulting in 80-90%
another lesion in the apical LAD, resulting in 80% stenosisThe doctor administered balloon angioplasty to the RCA stent. He then opened the LCX using three overlapping drug eluting stents and delivered another drug eluting stent to the LAD.
Figure D -
Although the patient was initially asymptomatic, the Guardian
device alarm prompted her to seek immediate medical treatment. At the hospital, conventional standard-of-care tests were negative and suggested no acute ischemia.
The Guardian device data however, showed two distinct acute ischemic events, with atypically elevated ischemia demonstrated over the course of about 90 minutes. These data persuaded the doctor to order catheterization, which revealed the multivascular stenosis and led to subsequent remedial treatment via PCI.
Angel Medical Systems, Inc.
1163 Shrewsbury Ave, Suite E
Shrewsbury, NJ 07702
Peripatetic Column Dick – Doc P Madhok* It is strange that the reproductive system are the days of submission and sublimation. It is a day of the aggressor and the sexualconquerer. Men want to be like Zorba, theGreek and women secretly desire the skillsAndrologist. And this is not talking about theof Cleopatra. Sexual jokes and conquests areopenly discussed in cocktail parties an
27 April 2012 Inaugural Seatrade Offshore Marine Asia (SOMA) Ends on a High The inaugural Seatrade Offshore Marine Asia (SOMA) conference, which came to a close at the Suntec Singapore International Convention and Exhibition Centre today, ended on a high as delegates participated in the final industry panel discussion. The "Technical and Environmental Challenges" session highligh